Xk3190-a12 weighing indicator hook up to scale. Simple_weighing_for_platform_scale
Only the indicator can transfer the data to computer, the computer can not control the indicator.
The 10, 20, 50 divisions are only valid when there is no decimal point. After the full capacity is reset, press the calibration switch at the back of the indicator to save the setting and get back to the weighing status.
Press the calibration switch at the back of the indicator to save the setting and get back to the weighing status Fast calibration of full capacity: Don t use it for canning system of compressive container.! XKA12 E indicator has the calibration protection and software tamper-proof function. Both the load cell and indicator are static sensitive devices.
Indicator will count according to last time sampling result. Indicator will count according to last time sampling result. In this process, if the Err 4 appears, it means sampling failed, the indicator will keep the result of last sampling. To protect the indicator and load cell, we cannot plug or withdraw the connector when the indicator is power on.
The format is as below: Weighing result including the decimal point When there is no decimal point, the 9th bit is blank 0x20 When the weight is negative, the 3rd bit is the negative sign 0x2d 10th and 11th byte: This is for relevant metrology bureau to test and control.
There is software version, software verifying code, calibration counter and parameter verifying code for legal metrology. A12E has no such button, on and off controlled by switch at the back 2.